J.Detect 9000 by DRC Techno
J.Detect 9000 is a desktop diamond verification instrument for loose and mounted diamonds. J.Detect 9000 is designed to separate synthetic diamonds from diamonds. J.Detect 9000 is manually operated but gives semi-automatic test interpretation. J.Detect 9000 is suitable for an environment where you need to test both mounted and loose diamonds e.g. dealers, diamond and jewellery manufacturers, retailers, buyers and gemmological laboratories. ASSURE ID: 24193
Size: 31.5 cm (L) x 27.5 cm (W) x 38 cm (H)
Weight: 14 kg
Price: 9,999 USD (August 2019)
What Stones Can Be Tested?
Weight of stones: 0.003 ct +
Size of stones: 0.7 mm +
Colour of stones: D-K
Shape of stones: All
Diamond simulants: No
Jewellery: Yes (please note this has not yet been tested by the ASSURE Program)
Capabilities according to DRC Techno
Automatically feed the stones? No
Automatically interpret the results? Semi-automatic
Automatically dispense the stones? No
Detect or refer synthetic diamonds? Refer
Multiple stones at one time? Yes
How Does it Operate?
The user manually places the loose or mounted stones inside the test chamber of the instrument. To start the testing, the user closes the test chamber and press the scan button.
J.Detect 9000 automatically makes a measurement and displays one of the following results:
In case of a query, J.Detect 9000 software has a troubleshoot option that connects the user with the DRC Techno Support team to get instant response, support team is available 24/6.
J.Detect 9000 was tested in accordance with the Diamond Verification Instrument Standard with the test protocol for instruments that separate diamonds from synthetic diamonds (Operation Category 1).
The tests took place at the independent laboratory UL in Canton, Massachusetts, United States.
J.Detect 9000 provides semi-automatic interpretation and the instrument was tested both with a novice and a manufacturer appointed expert. The manufacturer appointed an expert from DRC Techno. The below results are from the expert tests. Both the novice and the expert test results can be found in the summary report issued by UL. J.Detect 9000 was tested with the ASSURE Core Sample and ASSURE Smalls Sample.
The UL test report including more detailed test results can be found here.
ASSURE Core Sample
The ASSURE Core Sample consists of 1,000 loose diamonds and 200 loose synthetic diamonds. The stones are 0.02 ct to 0.20 ct in D-J colour. The below results are from the test with the ASSURE Core Sample.
|Diamond False Positive Rate||Ratio of synthetic diamonds incorrectly classified as 'diamond' to the total number of synthetic diamonds||14.6%|
|Diamond Referral Rate||Ratio of diamonds being referred for further testing to the total number of diamonds||18.6%|
|Diamond Accuracy||Ratio of diamonds correctly classified as ‘diamond’ to the total number of diamonds||81.4%|
|Synthetic Diamond Referral Rate||Ratio of synthetic diamonds being referred for further testing to the total number of synthetic diamonds||85.4%|
|Speed||Stones tested per hour||679|
ASSURE Smalls Sample
The ASSURE Smalls Sample consists of 1,000 loose diamonds and 200 loose synthetic diamonds. The stones are 0.005 ct to 0.02 ct in D-J colour. The below results are from the test with the ASSURE Smalls Sample.
|Diamond False Positive Rate||Ratio of synthetic diamonds incorrectly classified as 'diamond' to the total number of synthetic diamonds||3.7%|
|Diamond Referral Rate||Ratio of diamonds being referred for further testing to the total number of diamonds||9.7%|
|Diamond Accuracy||Ratio of diamonds correctly classified as ‘diamond’ to the total number of diamonds||90.3%|
|Synthetic Diamond Referral Rate||Ratio of synthetic diamonds being referred for further testing to the total number of synthetic diamonds||96.3%|
|Speed||Stones tested per hour||679|
Where Do I purchase the Instrument?
Phone: +91 990 9990 884
The most important requirement when selecting a Diamond Verification Instrument is that the instrument meets the safety regulations of your local jurisdiction. Before purchasing an instrument, request that the manufacturer verifies that their instrument complies with all regulatory requirements.
Click here to download the safety testing credential for J.Detect 9000 submitted by DRC Techno. Please note that the safety testing credentials have not been validated through the ASSURE Program.